Home › Environmental › Parameter Measurement › Ecom Instruments Ex-MX2 & Ex-MX4 BT Intrinsically-Safe Non-Contact Temperature Meter
Ecom Instruments Ex-MX2 & Ex-MX4 BT Intrinsically-Safe Non-Contact Temperature Meter
The Ex-MX2 is now obsolete and has been replaced by the FLUKE 568.
For non-contact temperature measurement and monitoring of temperature processes
in Ex-hazardous areas.
The Ex-MX2 and Ex-MX4 BT are the first
choice for monitoring temperatures on
objects which are difficult to access or for
moving materials. The precise and robust
measuring instruments measure the temperature
without contact, even in Zone 0.
Operation could not be easier: sight on
the object, press the measurement button
and read the measured value.
sighted measuring area can be very clearly
recognised using the 3-point measuring
area marking, and the device can be adjusted
and matched to the most diverse
materials using the emissivity setting (in
An exact measurement
result is obtained in this way. The device
has a 'High' (the Ex-MX4 BT also a 'Low')
alarm function which signals the exceeding
or falling below of a pre-set value
Only Ex-MX4 BT:
The values determined can be stored internally
(space for 100 measuring spots)
and called up on the display afterwards
by pressing a button.
Using the integrated Bluetooth function,
the data can be transmitted to an intrinsically
safe PDA directly in the ex-hazardous
area and/or to a PC with integrated Bluetooth
interface outside the ex-hazardous
The user can select settings such as
minimum and maximum value display
and difference and average value calculation
using the menu buttons.
If the emissivity of a material is unknown,
it can be determined using the included
measuring sensor. However, this function
is only allowed outside the ex-hazardous
Quick response time: 250 ms
3-point measuring area
Measure from Zone 1 into Zone 0
Only Ex-MX4 BT:
Data can be transmitted to a PC using
the integrated Bluetooth interface
Up to 100 measured values can be stored
Please note - Product designs and specifications are subject to change without notice. The user is responsible for determining the suitability of this product.
Back To Last Page