The low total cost of ownership and easy TOC analysis are just two of the things that make the Met One QbD1200 such a fantastic piece of equipment.
The setup process is now easier than ever, making the reliability of your results higher than ever. You no longer have to throw away your first replicate, as the QbD1200 has 95% less carryover.
The QbD1200 doesn't suffer fromi inconsistent results, with 2% standard deviation at 50 ppm and 3% at 100 ppb.
Users can enjoy annual servicing rather than monthly servicing, so you can wave goodbye to regular maintenance.
Calibration has also been made simple, with only 90 mins needed for a calibration routine, rather than all day.
Accuracy:± 2 %
Calibration Interval:1 Year; Time to Calibrate 90 Minutes
Calibration Method:Automated Routine: 18 Point Calibration Using KHP (6 Concentrations, 3 Replicates Each)
Carrier Gas Options:CO2 free Air, O2, or N2
Compliance:USP <643> (including Sterile Water SST), JP-16 <2.59>, EP <2.2.44>, IP, CP, KP, US EPA 5310c and 415.3
Data Export:PDF, CSV
Dimensions (H x W X D):410 mm x 320 mm x 507 mm
Display Type:10.4 inch Hi-Res Color Touch Screen
Inorganic Carbon Handling:No extra Inorgainc Carbon Removal Module needed
Overload Recovery:1 Measurement
Oxidation Method:UV Lamp + Persulfate
Power Requirements (Hz):47 - 63 Hz
Power Requirements (Voltage):100/240 VAC
Precision:3% or 3ppb, whichever is greater
Range:0.4 ppb - 100 ppm
Sample to Sample Carryover:<0.2%